We have the KYZ pulse output tied to our BMS in the building. We have a PLC set to count the pulses and calculate the energy consumption based on the pulses received but it doesn’t seem to match. What seems to be the problem? We use the factor 0.50001776 as shown on the meter label for TV < 150V which we assume the TV to be the voltage input. Our system is 120/208V system.
Electro Industries believes that quality is our MOST IMPORTANT company goal. To achieve that goal, we provide a multipoint approach to quality. EIG uses advanced standards, from the purchasing of components to the final shipping of products, to insure that our instruments are free from defects when shipped and operate for many years in service. We provide factory tours and have been know to impress people quite often.
In-Process Environmental Testing and 100% Inspection
EIG meters utilize brand-name, industrial temperature components. They receive 100% component insertion inspection, automated computer controlled testing, temperature deviation-based accelerated life burn-in testing and extensive 2.5 KV dielectric testing. We strive to achieve low failure rates by using comprehensive quality-based design and manufacturing techniques. We test extensively so that users can be confident that when they receive and EIG meter, it is one of the best in the industry.
How Can We Help?
Give us a call at 866-928-7908 or use this form.
[contact-form-7 404 "Not Found"]